Accutrend GC Blood Pressure Monitor pdf manual download. Since each code strip and test Glucose ® Accutrend, GC, GCT, DM CODE 50 TESTS. Description Reviews (0). ACCUTREND GCT PLUS BLOOD ANALYSIS METER. Test Report For: Glucose User Manual. Optional Request: In Exchange For. DownloadAccutrend gct user manual. Semoga Download driver keychain coby dp usb win xp, Hi osvaldorodriguez, are you If you are still a Windows user.
|Published (Last):||27 August 2016|
|PDF File Size:||13.68 Mb|
|ePub File Size:||13.37 Mb|
|Price:||Free* [*Free Regsitration Required]|
Code the meter see Section 3. Page 46 Move the meter to a different place see Section 5. If the strip code has not been read in correctly, error messages appear e. And if you are running a screening accuutrend, or just for use in your clinic, this is a tool for quick, reliable testing. Manufacturer’s Warranty And Guarantee For further information please ask your For disposal of the meter the rele- pharmacist or your Roche Diagnostics office.
October Softclix are trademarks of a Member of the Roche Group.
Comments to this Manuals Your Name. The round window on the back of the strip is not uniformly coloured. Accutend it up and make Nigerians proud. It is best to have your instrument next to you while reading the manual and to try out each step as it is described.
Got it, continue to print. Humidity In the presence of high humidity or large temperature changes condensation may form on the meter, e.
ROCHE ACCUTREND GC USER MANUAL Pdf Download.
This can lead to inappropriate treatment rec- accutrenr and thus endanger your health. The relevant regulations or guidelines applying to your institute or laboratory should be observed.
If at any time during the function test the display does not correspond to either of the diagrams on the left, please notify the Roche Diagnostics service see pages 97 and If the date and time have been set Section 2.
The optics are dirty or the meter is defective. Have new batteries ready. For further information on this Weight: Contact the Roche Diagnostics service centre. During the last four seconds a series of beeps is heard one beep per second followed by a longer beep at the end of the measurement. Get the results in a matter of seconds and minutes: The procedure for checking the measuring amnual is identical to the procedure for practising how to perform a measure- ment. Check the expiry date and storage conditions cgt the test strips.
With the GCT System, you easily do these tests on your own, at your convenience, with accurate, dependable results. Free Health Articles Select a topic to access it. If the instrument has not been coded before, the display shows. Clean the meter see Section 5. This is one major self-testing device to put you in good control of the direction of your health, even as you strive for the accomplishments you desire. Users in the office and laboratory sector should note that samples containing human material must be handled as potentially infectious material.
Don’t have an account?
Index Page Roche Diagnostics K. After 2 minutes with test strip inserted; after 1 minute if no button was pressed and no test strip has been inserted. System Components For further information please ask your For disposal of the meter the rele- pharmacist or your Roche Diagnostics office.
Table of Contents Add to my manuals Add. The used test strips and consumables e. It should also be cleaned: On target high relevance: The Accutrend GCT System is most suitable for accurate on-site screening, diagnosis, and therapy monitoring of metabolic disorders and cardiovascular risk factors.
Accktrend of manuao Go.
Roche Accutrend GC User Manual
The time must also be changed from summer to win- ter time and vice versa, and 29 February must be entered in a leap year. Recode the meter with the code strip from the pack of test strips you are using see Section 3.
Nanual the relevant directory by pressing for glucose values or for cholesterol values. By pressing ‘print’ button you will print only current page. If other messages appear on the display, consult Sec- Too little blood applied — If any of these items are missing or damaged please contact your supplier.
Ready for setting the date and time or the beeper. It’s designed for use whether in the physicians’ office or at home by patients. If the problem persists, contact the Roche Diagnostics service centre see Section 6. Keep pressing to set the minutes in the same way as the hour. When the test CODE strip is inserted the display automatically switches to the cor- rect symbol for the strip.